The instrument shall not be used for purposes different from those described hereunder. It cannot be used as a safety device. Check the application limits before proceeding. Do not expose to water or moisture: use the controller only within the operating limits avoiding sudden temperature changes with high atmospheric humidity to prevent formation of condensation Warning: disconnect all electrical connections before any kind of maintenance.
|Published (Last):||22 September 2010|
|PDF File Size:||14.34 Mb|
|ePub File Size:||12.15 Mb|
|Price:||Free* [*Free Regsitration Required]|
The instrument shall not be used for purposes different from those described hereunder. It cannot be used as a safety device. Check the application limits before proceeding.
Do not expose to water or moisture: use the controller only within the operating limits avoiding sudden temperature changes with high atmospheric humidity to prevent formation of condensation Warning: disconnect all electrical connections before any kind of maintenance.
Fit the probe where it is not accessible by the End User. The instrument must not be opened. In case of failure or faulty operation send the instrument back to the distributor or to "Dixell S. Consider the maximum current which can be applied to each relay see Technical Data. Ensure that the wires for probes, loads and the power supply are separated and far enough from each other, without crossing or intertwining.
In case of applications in industrial environments, the use of mains filters our mod. FT1 in parallel with inductive loads could be useful. They are provided with six relay outputs to control compressor, defrost - which can be either electrical or hot gas - the evaporator fans, the lights, the alarm and an auxiliary output.
In XWL the auxiliary output is configured as anti-condensing heater. They are also provided with three NTC probe inputs, one for temperature control, one to control the defrost end temperature of the evaporator and the third, optional, for the display. There are two digital inputs free contact for the door switch and configurable by parameter.
In case of fault in the thermostat probe the start and stop of the compressor are timed through parameters "COn" and "COF". The compressor operates in continuous mode for the time set through the "CCt" parameter. The cycle can be terminated before the end of the set time using the same activation key, 3.
At the end of defrost the drip time is controlled through the "Fdt" parameter. This can be used to make sure circulation of air only if his temperature is lower than set in "FSt". The auxiliary output of the XWL model controls the anti-condensing heater and it is automatically activated if the room temperature is lower than the "SAA" parameter. By holding it pressed for 3s when max or min temperature is displayed it will be erased.
By holding it pressed for 3s the fast freezing cycle is started. To see the min stored temperature; in programming mode it browses the parameter codes or decreases the displayed value. By holding it pressed for 3s the defrost is started. By holding it pressed for 3s Energy Saving function is started or stopped. For XWL model.
CONTRÔLEUR ÉLECTRONIQUE DIXELL XW260L-5N0C5
This time allows the evaporator to eliminate water drops that. ALU High temperature alarm setting:. FSt Fan stop temperature: Learn more — opens in new window or tab. Buy it now — Add to Watch list Added to your Watch list.
XW260L Dixell 230V 20A Electronic temperature controller
Gozahn Got it, continue to print. To print the manual completely, please, download it. Parcels sent out on the standard next day courier can be delivered any time between 7: If nobody is available then cixell parcel will be returned to us. Learn more — opens in new window or tab. The seller will not accept any responsibility for any losses incurred as a direct or indirect result of any parts or equipment supplied including delayed delivery.
Elektronische regelaar Dixell